×
INTELLIGENT WORK FORUMS
FOR ENGINEERING PROFESSIONALS

Are you an
Engineering professional?
Join Eng-Tips Forums!
• Talk With Other Members
• Be Notified Of Responses
• Keyword Search
Favorite Forums
• Automated Signatures
• Best Of All, It's Free!

*Eng-Tips's functionality depends on members receiving e-mail. By joining you are opting in to receive e-mail.

#### Posting Guidelines

Promoting, selling, recruiting, coursework and thesis posting is forbidden.

# Nozzles, temperature, pressure

## Nozzles, temperature, pressure

(OP)
Hello There
Hope you are fine.

I hope to be posting my question in the right forum

I am utilizing some calibrated nozzles in order to calculate the air flow on an equipment.
These nozzles are subsonic nozzles therefore governing equation is

Qm = C/(1-β^4)*ε*3.14/4*d^2*(2*DP*rho1) ^0.5 (1) kg/s
Where
d is the nozzle outlet diameter
DP is the pressure difference in the nozzle
Rho1 is the air density at the nozzle inlet
Other symbols are nozzle characteristics that won’t change for the scope of the question.

These nozzles have been provided with a calibration certificate.
The certificate provides me a mass flow (let’s call Qm) at calibration conditions, let’s say Tcal and Pcal (temperature and pressure at calibration time) , in function of different pressure differences.

Following is my question.
Let’s assume we have in test different conditions at the nozzle inlet, said Ttest and Ptest (temperature and pressure at nozzle inlet in test condition)
How can I convert the calibrated values to the test conditions ?
1) I should simply apply mass conservation therefore

Volume flow (Ttest,Ptest) = Qm/rho(Ttest,Ptest) m³/s

2) Since Pcal is different from Ptest, and Tcal is different from Ptest I need
2.1 - First to apply the correction on density at the inlet of the nozzle, since Qm is derived from conditions Tcal and Pcal instead than Ttest and Ptest, said therefore since the nozzle is the same the (1) becomes

Q2 = Qm_cal*(rho_cal/rho_test)^0.5

And then
2.2)

Volume flow (Ttest,Ptest)= Qm2/rho(TB,PB)
What changes with respect to 1) is that now we apply a further correction due to inlet conditions different from calibration

Stay safe

### RE: Nozzles, temperature, pressure

For the calibrated nozzle used in the testing, should the Vendor provide a conversion table of method to be used to calculate the real mass /volume flow rate?
Without knowing the detail of certificate testing nozzle, it's difficult to figure it out the real answer.

### RE: Nozzles, temperature, pressure

Have you reviewed this issue with the supplier of the nozzles?

### RE: Nozzles, temperature, pressure

(OP)
Hello

@mk32323
The vendor is providing only following data
mass flow
DP
temperature
relative humidity
pressure
(therefore density)
in calibration conditions

@bimr
Unfortunately the supplier of nozzles can't help me since he asserts of not knowing in detail how to change from calibration data to other conditions.

#### Red Flag This Post

Please let us know here why this post is inappropriate. Reasons such as off-topic, duplicates, flames, illegal, vulgar, or students posting their homework.

#### Red Flag Submitted

Thank you for helping keep Eng-Tips Forums free from inappropriate posts.
The Eng-Tips staff will check this out and take appropriate action.

#### Resources

Taking Control of Engineering Documents
This ebook covers tips for creating and managing workflows, security best practices and protection of intellectual property, Cloud vs. on-premise software solutions, CAD file management, compliance, and more. Download Now
The Great Project Profitability Debate
A/E firms have a great opportunity to lead the world into the future, but the industryâ€™s greatest assetâ€”real-time dataâ€”is sitting wasted in clunky, archaic ERP platforms. Learn how real-time, fully interactive dashboards in a modern ERP allow you to unlock data that will shape the future of the world. Download Now

Close Box

# Join Eng-Tips® Today!

Join your peers on the Internet's largest technical engineering professional community.
It's easy to join and it's free.

Here's Why Members Love Eng-Tips Forums:

• Talk To Other Members
• Notification Of Responses To Questions
• Favorite Forums One Click Access
• Keyword Search Of All Posts, And More...

Register now while it's still free!