×
INTELLIGENT WORK FORUMS
FOR ENGINEERING PROFESSIONALS

Are you an
Engineering professional?
Join Eng-Tips Forums!
• Talk With Other Members
• Be Notified Of Responses
• Keyword Search
Favorite Forums
• Automated Signatures
• Best Of All, It's Free!

*Eng-Tips's functionality depends on members receiving e-mail. By joining you are opting in to receive e-mail.

#### Posting Guidelines

Promoting, selling, recruiting, coursework and thesis posting is forbidden.

# Blower Flow rate measurement

 Forum Search FAQs Links MVPs

## Blower Flow rate measurement

(OP)
Hi All!

Can anyone help us in finding out what would be the best possible location of a pitot tube for the flow measurement in a blower?

Currently, we are locating it at the discharge line, but we have seen some arrangements showing it at the suction of the blower. Is there any significant difference??

Thx,

ChemE2912

### RE: Blower Flow rate measurement

Hi,

Maybe it's better to ask this question on the measurement & control instrumentation group.

> There's no principal difference to measure in the suction or in the discharge.

1) Keep in mind to measure not to close to bends or other obstacles, because turbulence will spoil your measurement then.

2) Then what exactly do you want to measure? Volumeflow (m3/sec) or massflow (kg/sec). For massflow you need to measure pressure and temperature as well since the density of gasses change with pressure and temperature.

3) At what point are you going to measure? In the centre of the channel the flow will be higher then near the wall.

There are good Pitot systems around who take say 5 measurements across the whole channel and take the average.

See for example Endress + Hauser: Deltatop / Deltaset.

http://www.endress.com/

These are complete systems with flow computer, where all temperature and pressure caculations are done to covert to massflow.

Hope this helps,
MVD

### RE: Blower Flow rate measurement

You r measuring at the S/C r D/C,  the flow will remain same,since you have to use correct density in the formula
V=sqrt(2GH),Where H is the FT of air column. H is obtained by multiplying the Press by density of manameter fluid and dividing Press by Density of air for the corresponding press where pitot is inserted.

### RE: Blower Flow rate measurement

Suction flow measurement of an (atmospheric) air blower is the total flow of the blower. Discharge flow could be either the total or the net flow depending on the flow location relative to the location of the blow off line/valve. So, comparing total in suction to net in discharge would/could be significant. Do you want to measure net or total?

Averaging Pitot tubes are prefered to pitot tube (as MVD points out). Annubar is the name used by Rosemount.

Discharge lines typically have less diameter and often has the advantage of more choice in selecting the location. Another advantage is that pressure drop in discharge is less of a loss (operationally cheaper) than in the suction (dP/P). DPs in suction of blower, for pitot tube, are probably very small (draft range or less) which can be problematic.

dont forget to mount dp instrument above the taps with no utubes, condensed and trapped h2o will create alot of problems.

only observed pitot tubes in cheme lab, never in industry.

### RE: Blower Flow rate measurement

Hi ChemE2912,

drop your e-mail adress and I'll sent you two nice spreadsheets on air flow mass balance across an air blower.

Cheers,
MVD

### RE: Blower Flow rate measurement

done!

Good luck,
MVD

#### Red Flag This Post

Please let us know here why this post is inappropriate. Reasons such as off-topic, duplicates, flames, illegal, vulgar, or students posting their homework.

#### Red Flag Submitted

Thank you for helping keep Eng-Tips Forums free from inappropriate posts.
The Eng-Tips staff will check this out and take appropriate action.

#### Posting in the Eng-Tips forums is a member-only feature.

Click Here to join Eng-Tips and talk with other members! Already a Member? Login

#### Resources

eBook - Functional Prototyping Using Metal 3D Printing
Functional prototypes are a key step in product development â€“ they give engineers a chance to test new ideas and designs while also revealing how the product will stand up to real-world use. And when it comes to functional prototypes, 3D printing is rewriting the rules of whatâ€™s possible. Download Now

Close Box

# Join Eng-Tips® Today!

Join your peers on the Internet's largest technical engineering professional community.
It's easy to join and it's free.

Here's Why Members Love Eng-Tips Forums:

• Talk To Other Members
• Notification Of Responses To Questions
• Favorite Forums One Click Access
• Keyword Search Of All Posts, And More...

Register now while it's still free!