×
INTELLIGENT WORK FORUMS
FOR ENGINEERING PROFESSIONALS

Are you an
Engineering professional?
Join Eng-Tips Forums!
• Talk With Other Members
• Be Notified Of Responses
• Keyword Search
Favorite Forums
• Automated Signatures
• Best Of All, It's Free!

*Eng-Tips's functionality depends on members receiving e-mail. By joining you are opting in to receive e-mail.

#### Posting Guidelines

Promoting, selling, recruiting, coursework and thesis posting is forbidden.

# Measure flux density of NdFeb magnet

## Measure flux density of NdFeb magnet

(OP)
I would like to measure the quality of NdFeb.

Following is the specification of the magnet

Dimension: 5 x 5 x 2 mm
Magnetization: 4 poles on each 5x5 mm^2 surface. (4 equal quadrants surface... like rice field)

I would like to determine magnetization quality like the
- Flux density
- "airgap" dimension separating N..S..N..S. The width of the line is about <0.2mm.

Can anyone suggest a simple method to measure these parameter?

### RE: Measure flux density of NdFeb magnet

what equipment do you have available?

### RE: Measure flux density of NdFeb magnet

(OP)
I have a Gaussmeter meter (GM05 by Hirst), many hall elements.

### RE: Measure flux density of NdFeb magnet

I believe the correct way is to design a fixture with an closed flux path (soft iron yoke, for example) with a small air gap where you perform the measurement. it might be more difficult than it is worth because of the geometry of your part. the field in the gap to the "net" flux conducted through the measurement yoke.  the big question is making certain that the "net" flux is representative of the material.

meas. of the field next to the pole face without a yoke, is going to be a sum of the fields produced by all adjacent magnetized surfaces. it will scale (be proportional to) the number you are after but it won't be absolute because of the geometrical factors.

force measurements (lift-off force from soft yoke) that are possible, but they too are indirect and require a lot of work to develop calibration factors.

one thing that you can do is to develop a "go by test" wherein you measurement "flawed" though it may be is performed consistently. once you quantify the readings for a good part, you hold all others to that "standard".

good luck

#### Red Flag This Post

Please let us know here why this post is inappropriate. Reasons such as off-topic, duplicates, flames, illegal, vulgar, or students posting their homework.

#### Red Flag Submitted

Thank you for helping keep Eng-Tips Forums free from inappropriate posts.
The Eng-Tips staff will check this out and take appropriate action.

#### Resources

Low-Volume Rapid Injection Molding With 3D Printed Molds
Learn methods and guidelines for using stereolithography (SLA) 3D printed molds in the injection molding process to lower costs and lead time. Discover how this hybrid manufacturing process enables on-demand mold fabrication to quickly produce small batches of thermoplastic parts. Download Now
Examine how the principles of DfAM upend many of the long-standing rules around manufacturability - allowing engineers and designers to place a partâ€™s function at the center of their design considerations. Download Now
Taking Control of Engineering Documents
This ebook covers tips for creating and managing workflows, security best practices and protection of intellectual property, Cloud vs. on-premise software solutions, CAD file management, compliance, and more. Download Now

Close Box

# Join Eng-Tips® Today!

Join your peers on the Internet's largest technical engineering professional community.
It's easy to join and it's free.

Here's Why Members Love Eng-Tips Forums:

• Talk To Other Members
• Notification Of Responses To Questions
• Favorite Forums One Click Access
• Keyword Search Of All Posts, And More...

Register now while it's still free!