×
INTELLIGENT WORK FORUMS
FOR ENGINEERING PROFESSIONALS

Are you an
Engineering professional?
Join Eng-Tips Forums!
• Talk With Other Members
• Be Notified Of Responses
• Keyword Search
Favorite Forums
• Automated Signatures
• Best Of All, It's Free!

*Eng-Tips's functionality depends on members receiving e-mail. By joining you are opting in to receive e-mail.

#### Posting Guidelines

Promoting, selling, recruiting, coursework and thesis posting is forbidden.

(OP)
hello everyone. I have one question regarding the RTD Platinum PT100 output.

we have 3 RTD install on heat exchanger after replacing shell and tube reading reduces from 117 to 111

As per our process indication and energy consumption the Temp Transmitters it actually should read higher than current reading.

What will be the possible cause for temperature reduce while process temperature should be around 118 C.

There are two standards.
One has a 0 DegC resistance of 100 Ohms, the DIN standard calls for 98.129 Ohms

If DIN RTD's replaces the first, then 118 Deg C would read cooler say 115 Deg C

The error difference for a 392 alpha vs a 385 alpha RTD at 117°C is only about 2°C, so the 6 degree C error (117 vs 111) comes from somewhere else.

The 392 alpha RTD curve is the standard for SAMA (American Scientific Apparatus Makers Association) RC21-4-1966 and the US Department of Defense (MIL-T-24388). I've heard people refer to the 392 alpha as the Burns Engineering alpha, because Burns was a major proponent and vendor for such. The Asians frequently refer to the 392 alpha as the American standard.

I have to say that I haven't run into 392 alpha RTD's since the 1990's, everything I've run into in the past 20 years has been DIN 385 RTDs in the US, but that might be a limitation of my industrial social neighborhood;

The 60751 DIN standard is the 385 alpha.

The instrument interpreting the voltage created by the current source driven through the RTD resistance has to match the RTD alpha curve.

If the instrument is configured for the 'other' alpha' curve, then yes, there will be an error:
- a 392 alpha instrument reads low with 385 RTD.
- a 385 alpha instrument reads high with 392 RTD.

Specifically, for a 392 alpha instrument reading low with a 385 alpha RTD:
At 117.0°C, a 392 alpha instrument expects 145.75 Ohms, sees only 144.937 Ohms from a 385 alpha RTD;
the 392 instrument reports a temperature of 114.9°C, a low error of 2.1°C

2.1°C is 1/3 of the 6 degree error reported.

#### Red Flag This Post

Please let us know here why this post is inappropriate. Reasons such as off-topic, duplicates, flames, illegal, vulgar, or students posting their homework.

#### Red Flag Submitted

Thank you for helping keep Eng-Tips Forums free from inappropriate posts.
The Eng-Tips staff will check this out and take appropriate action.

#### Resources

Low-Volume Rapid Injection Molding With 3D Printed Molds
Learn methods and guidelines for using stereolithography (SLA) 3D printed molds in the injection molding process to lower costs and lead time. Discover how this hybrid manufacturing process enables on-demand mold fabrication to quickly produce small batches of thermoplastic parts. Download Now
Examine how the principles of DfAM upend many of the long-standing rules around manufacturability - allowing engineers and designers to place a partâ€™s function at the center of their design considerations. Download Now
Taking Control of Engineering Documents
This ebook covers tips for creating and managing workflows, security best practices and protection of intellectual property, Cloud vs. on-premise software solutions, CAD file management, compliance, and more. Download Now

Close Box

# Join Eng-Tips® Today!

Join your peers on the Internet's largest technical engineering professional community.
It's easy to join and it's free.

Here's Why Members Love Eng-Tips Forums:

• Talk To Other Members
• Notification Of Responses To Questions
• Favorite Forums One Click Access
• Keyword Search Of All Posts, And More...

Register now while it's still free!