we have a rosemount 3051s differential flow transmitter mounted on a RO SKID.As part of calibration we checked the instrument before installation and found the transmitter showing very large error in zero(420mbar)when the calibrate range in 0 to 250mbar.so we tried to do zero trim the instrument from field communicator but it failed and showing that excess correction.anyone know how to solve this problem? waiting for your help...
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
Manufacturers are at a critical juncture. Either they evolve their product processes or get disrupted by new competitors who have been built from the ground up to develop complex systems and products defined by software. Download Now
Savvy engineers have eagerly anticipated the impact of virtual reality (VR) on robot programming technology, as it can dramatically increase productivity by empowering engineers to do their jobs safer and more efficiently. Download Now