How we can measure the our FF signal .means if i am calibrating a PT honeywell 800 how can i measure my output iscorrexct to the input .e.g in hart if my PT is 0 to 100bar i have give input 50bar i can measure 12mA output and in FF how i can measure with out any communicater e.g hart 475
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.