If I have ultrsoinc meter for gas and the flow range are (320-10000)m3/h and I calibration in the lab from (320-6400)m3/h. So is that ok or not and why. Because in AGA9 report recommended to do calibration from min.to max.
Can help me plz
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
PCB design is a purposeful process. In essence, every PCB design is a complex puzzle that requires a designer to incorporate critical components and tight schematics into a piece of technology that is governed by rules, constraints and industry standards. Download Now