I have a problem with our thermocouples jumping the temperature readings either extremely high or low as soon as we apply a current to the test loop. Once the current is shut off the TC's go back to normal measurements. I was at a lab and they had grounded a part of their transformer to avoid this problem. Does anyone have any idea what to ground or how to fix this problem?
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
By implementing a new 3D CAD tool, top-performing manufacturers can reduce their development time by 19 percent and development costs by 15 percent. Most importantly, engineers can increase iterations by 14 percent â€” leading to more innovation and competitive advantages. Download Now