## PROVING RING CONSTANT

## PROVING RING CONSTANT

(OP)

HOW TO FIND PROVING RING CONSTANT, AS I READ MANY POST THEN I COME TO KNOW THAT BY PLOTTING GRAPH IN BETWEEN STANDARD LOAD TO THE READING OF PROVING RING, AFTER THAT FIND THE SLOPE OF THAT LINE, SO MY ANOTHER QUOTATION IS WHICH VALUE SHOULD BE ON X-AXIS AND Y- AXIS PLEASE HELP ME BECAUSE WE WILL GET 2 DIFFERENT SLOPES WHEN WE WILL INTERCHANGE THE VALUE

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

Column A has load numbers.

Column B Has dial readings corresponding to the load numbers.

Column C has corrected dial readings by subtracting initial dial reading from Col. B.

Column D divides Col A by Col C

At the bottom of the data entries compute the average of values in Col. D as the calibration factor for the proving ring.

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT

Also, you should not have a "Y" axis intercept, even with your trend curve (straight line). Zero load is Zero deflection.....

## RE: PROVING RING CONSTANT

## RE: PROVING RING CONSTANT