I noticed the thread "Factors affecting the viscosity of palm oil" was closed before I had a chance to comment. The problem posed was wildly different viscosity readings. The list of factors affecting viscosity was a good list. But what I would have added was the measurement itself. Brookfield viscosity can be very sensitive to temperature and human factors in testing, especially with complex natural chemicals in the food industry. At one place I worked, a team performed an MSA on viscosity measurement of starch-based and flour-based batter. Results showed that a viscosity cup measurement (Accu Dyne is one vendor)gave a more accurate and meaningful reading than Brookfield viscosity. I have seen Brookfield results range from 200 to over 1000 cps over a shift when the cup viscosity measurements showed there was little change during the shift. I don't know which instrument was used to measure viscosity for that closed thread I mentioned, but the fluctuating viscosity results caught my attention. If anyone else has experience with viscosity measurement that could either support, refute or clarify my observations, please respond.
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
Cities are ceaseless engines of engineering activity. Itâ€™s likely that while you were on your way to work, you encountered a number of infrastructure development projects under construction. Download Now
When was the last time you drove down the highway without seeing a commercial truck hauling goods?The ubiquity of these machines underscores the important roll they play in modern commerce. Download Now