When calibrating the Type B Air Meter, if the Initial Pressure (IP) Line is higher than 5, can the gauge be reset or does it have to be replaced? The IP line fluctuates with seemingly every calibration, which is performed every three months, and sometimes it goes beyond 5, which is not recordable. Normally I would replace the gauge.
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
Since the later 1980s, manufacturing companies have seen over 3 percent in labor output year on year, with most of this productivty enhancement occuring because of the optimization of repetitive tasks. Download Now