Last week, our Nikon DTM-A5 total station began to malfunction. When shooting distance, the instrument will go through its "system warmup", begin to measure the distance, time out, and retry. I have changed out the prisms, adjusted the angle of the prism, charged batteries, and tried with and without the data collector. Has anyone had any similar problems or suggestions (besides get rid of the antique)? Thanks for any help.
Red Flag Submitted
Thank you for helping keep Eng-Tips Forums free from inappropriate posts. The Eng-Tips staff will check this out and take appropriate action.
Reply To This Thread
Posting in the Eng-Tips forums is a member-only feature.
Since the later 1980s, manufacturing companies have seen over 3 percent in labor output year on year, with most of this productivty enhancement occuring because of the optimization of repetitive tasks. Download Now